PD300R-UV — это круглый фотодиодный датчик с широким спектральным диапазоном для измерения лазера. Он имеет апертуру 10 мм и съемный фильтр. Без фильтра его спектральный диапазон: 200—1100 нм, а диапазон измерения мощности: от 20 пВт до 3 мВт. С фильтром его спектральный диапазон: 220—1100 нм, а диапазон измерения мощности: от 2 мкВт до 300 мВт. Датчик поставляется с 1,5-метровым кабелем для по
- Ø35 W x 24 D (mm)
- 0.2 s
- CE, China RoHS
What does the PD300 "Background Subtraction" do?
Ophir's Photodiode PD300 and PD300-1W sensors offer automatic background subtraction so the measurement is not sensitive to room light. With "filter out" (i.e. the external filter removed for low light measurements), 2 separate detector elements are visible. The beam to be measured is incident only on the outer of the 2 detectors, but background light reaches both detectors. The instrument will show the power measured by the outer detector minus that measured by the inner detector.This patented method cancels out 95% - 98% of background light under normal room conditions, even if it is constantly changing.Close
How do I take off the removable filter from the PD300?
Can I use a sensor from the PD300 family to measure average power of pulsed lasers?
In general yes, but several technical issues need to be kept in mind (most of which are results of the fast physical response time of these sensors):
- The pulse rate should be more than about 30Hz, otherwise the reading is unstable. At higher pulse frequencies, the sensor will respond as if the beam were CW.
- It is possible for a pulsed beam to have average power within the sensor spec and yet have the energy of the pulses themselves be high enough to cause a momentary saturation of the sensor. It is important to be sure that pulse energy is also within sensor spec (the parameter "Max pulse energy" is included in all specs for the PD300 family, for just this reason).
- The beam diameter should be no less than about 1mm .
- The average power and power density restriction in the spec should not be exceeded
Note: At the maximum pulse energy limit given in the spec, the reading will be saturated by about 5%, i.e. the reading will be about 5% lower than it should be. At 1/3 the maximum, the saturation will be about 1%.Close
Can a lost PD300 filter be replaced?
Technically it could be replaced, but it is not just a matter of replacing the filter. Since the PD300 is a "calibrated" sensor it requires that the filter also be "calibrated". Especially since the PD300 response varies with wavelength, it requires that both the PD300 and the filter be calibrated over the entire spectral range with a monochromator. Because of the cost to calibrate the replacement filter with the PD300 sensor, we recommend purchasing a new PD300 sensor when a replacement filter is needed.Close
When using a photodiode laser power sensor to measure very low power pulsed beams (nW to mW), there are some issues you need to be aware of. This video shows you how to avoid some common problems and ensure maximum accuracy.
Are you measuring a laser beam coming at an angle correctly?
Ever wonder how your laser power measurements are affected by your beam’s angle of incidence?
In this video, you will learn to what degree a beam’s incidence angle matters, and for which sensor types this should be taken into account.
In this short “Basics” video, we review in general the use of photodiode sensors for measuring very low laser powers.
Laser Power Sensors introduction
As described in the general introduction, the thermopile sensor has a series of bimetallic junctions. A temperature difference between any two junctions causes a voltage to be formed between the two junctions. Since the junctions are in series and the «hot» junctions are always on the inner, hotter side, and the «cold» junctions are on the outer, cooler side, radial heat flow on the disc causes a voltage proportional to the power input. Laser power impinges on the center of the thermopile sensor disk (on the reverse side of the thermopile), flows radially and is cooled on the periphery. The array of thermocouples measures the temperature gradient, which is proportional to the incident or absorbed power. In principle, the reading is not dependent on the ambient temperature since only the temperature difference affects the voltage generated and the voltage difference depends only on the heat flow, not on the ambient temperature. Прочитать подробнее…
Common Reasons for Photodiode Sensor Damage or Out of Tolerance Conditions
We have included this document with your recent calibration order because we have noticed an out of tolerance condition obtained from your equipment when returned for calibration. This document was created to assist our valued customers in the proper care and maintenance of Ophir photodiode sensors. The following information is for reference only. If you have any reason to believe that the sensor is no longer performing within the original specifications, we always recommend that you send it in for repair and/or recalibration by our trained technicians to bring the unit back to the proper NIST traceable standards.
Ophir photodiode sensors can be used for many years without any repairs when used with the proper laser optical setup. Many of our customers have sensors that are using their original absorber that are over ten years of age. We hope that this document will enable you to also enjoy the long life and reliable results that Ophir- Spiricon is known for. Прочитать подробнее…
Effect of Ambient Conditions on Laser Measurements
We are often asked about the specified ranges of various ambient conditions (temperature, humidity, etc.) for Ophir instruments. In this article we will clarify the effects of these conditions on laser measurements, so you’ll be able to use your Ophir laser measurement instrument effectively. Прочитать подробнее…
Online Sensor Finder
- Ø35 W x 24 D (mm)
- 0.2 s
- CE, China RoHS
Ø7mm aperture CDRH adapter for PD300R series
(Dimensions: Ø35 W x 9.2 D (mm))
7Z08227This fiber adapter is used for connecting power and energy sensors to a standard SC-type fiber. Many sensors need an additional mounting bracket to connect to all fiber adapters. More information can be found in the datasheet below.
7Z08226This fiber adapter is used for connecting power and energy sensors to a standard ST-type fiber. Many sensors need an additional mounting bracket to connect to all fiber adapters. More information can be found in the datasheet below.
7Z08229This fiber adapter is used for connecting power and energy sensors to a standard FC-type fiber. Many sensors need an additional mounting bracket to connect to all fiber adapters. More information can be found in the datasheet below.
1G01236This fiber adapter is used for connecting power and energy sensors to a standard SMA-type fiber. Many sensors need an additional mounting bracket to connect to all fiber adapters. More information can be found in the datasheet below.
1G02259A mounting bracket is needed to connect round photodiode sensors to a fiber adapter (SC, ST, FC or SMA). This bracket can be used for the PD300R (round) photodiode series, as well as the FPS-1 fast photodiode. It is not needed for the PD300-IRG sensor.
For mounting to SM1 optical components and systems