By Dick Rieley, East Coast Regional Sales Manager, Ophir Photonics Group
The U.S. military often serves as the proving grounds for emerging technologies. With advances in laser technologies, it is more common today for these military systems to involve lasers. One such example is the use of a laser in the periscope of a nuclear-powered submarine. As you can imagine, the data that this laser provides relies directly on both the quality of the laser and its alignment within the system. So how is the accuracy of this particular system’s laser ensured?
Which Measurement Method
The data collected from the measurement of these lasers must be the most accurate that the industry can supply. To supply this data, Ophir-Spiricon BeamGage® beam diagnostic software was used along with the Spiricon SP300, a high-speed, high-resolution USB3 CCD camera. It offers a high resolution array, 7.1mm x 5.3mm, and pixel spacing of 3.69um x 3.69um. An active area of 1928 x 1448 pixels was used.
The challenge of this beam measurement application was the identification and location of the center of a large area laser alignment beam, <30mm. Due to the size of the beam and the >30M distance to the target plane, the application did not lend itself to locating an exacting position and alignment of the beam. Several attempts at this alignment using IR sensitive sensors and materials did not prove sufficient to the accurate positioning of the beam. The solution was eventually obtained through the use of the Spiricon SP300, large area USB CCD camera, operating with BeamGage Professional software.
Figure 1: 2D & 3D Beam Profiles of Laser’s Raw Beam Ellipticity: 97% | Gaussian Goodness of Fit: 90%
Three key measurements were made: 1) a baseline measurement of the source beam using the 1/e2 measurement value, 2) a measurement calculation of the beam’s ellipticity (roundness), and 3) a calculation as to the laser source’s Gaussian, or Goodness of Fit value.
The analysis of this laser source was first conducted with the SP300 camera positioned at 1m from the source, without optics, splitters, or other optical interference between the source and the CCD array. The camera was only installed with an ND2 attenuator to avoid the camera saturating, a condition that can, if allowed to persist, damage the array and produce meaningless measurement values.
Critical Measurements on the Raw Beam
The first beam analysis was conducted on the raw beam from the laser source to establish a baseline of the size and shape of the beam, the orientation of the beam (round or oval), and, if oval, the direction of the minor and major axis positions. Due to the critical nature of the quality of the laser source, having a beam that possesses an ellipticity value of 92% or better is essential (a perfectly round beam has an ellipticity of 100%). Most high quality laser sources demonstrate an ellipticity value of 90% or better. In most high reliability applications, any measurement that is within 10% of an absolutely perfectly round beam meets specification.
The beam size, measured using the 1/e2 (13.5% of Peak) technique as specified by the manufacturer of the source, was validated by the BeamGage measurement when imaged at a short distance from the source. The overall shape of the beam and the distribution of the intensity internal to the beam was another issue that had yet to be diagnosed.
Once the overall beam size had been validated and determined to be within specification, the next analysis required the measurement of the ellipticity (or roundness) of the beam. BeamGage Professional software offers this quantitative measurement and graphic validation of this condition by activating the ellipticity measurement, which is a ratio of the major and minor axes. The quantitative measurement of ellipticity was determined to be 97%. Figure 1 shows the major crosshair (from lower left to upper right) and the minor crosshair (from lower right to upper left). The actual difference in the beam diameter measurement vs the same measurement when the ellipticity measurement is activated is only a difference of 197um.
Critical Measurements at a Distance
The last procedure in the analysis of the laser source involved repeating these measurements when the beam profiling system is positioned at greater distances from the source. This allows for determination of which of these measurements change, the degree of change, and if those changes are within tolerance.
Two additional sets of measurements were initially taken, each at an increased distance from the source. The purpose was to determine changes in beam diameter, ellipticity values, and Goodness of Fit values.
At this intermediate position, change had occurred; the beam size had increased by 5%, which means that the beam had some divergence present. Also, the ellipticity had deteriorated by 16% while the goodness of fit remained largely unchanged.
What Was Learned