The Pyro-C sensors have a "user threshold" feature allowing the user to adjust the measurement threshold in noisy environments. Increasing the threshold will prevent triggering on noisy signals and allow accurate measurment of energy and frequency, as long as the laser pulses are larger than the noise.
The trigger level can be adjusted up to 25% of full scale, however operation depends on the pulse width setting. For pulse width settings below ~0.25ms, the minimum energy that can be measured accurately is approximately 40% above the user threshold setting. Pulses below this energy level will trigger the sensor down to the user threshold level, but accuracy is compromised.
For pulse width settings above ~0.25ms, accuracy is good all the way down to the threshold. If the laser pulse width is less than 1/2 the setting, the minimum energy corresponds to the setting. However, with longer laser pulse widths, the minimum energy will be higher, rising to approximately twice the user threshold level when the laser pulse width is equal to the sensor pulse width setting.
It is recommended always to set the user threshold to the minimum possible setting to retain best energy accuracy in any given situation. See the user manual for further information on how to use the user threshold.