Камера профилометра SP928
Камера SP928 точно фиксирует и анализирует длины волн 190—1100 нм. Она отличается компактностью, широким динамическим диапазоном, беспрецедентным отношением сигнал/шум и встроенной цепью предварительного срабатывания. Благодаря этому камера идеально подходит для измерения профилей лазеров непрерывного излучения и импульсных лазеров.
- Разрешение 1928 x 1448 пикселей с шагом пикселя
- 37μm - 5.3mm
- USB 3.0
- Silicon CCD
- CW, Pulsed
- 5.3mm x 7.1mm
- 1928 x 1448
- 13 fps
- CE, China RoHS
BeamMic software, software license, 1/1.8” format 1928X1448 pixel camera with 4.5mm CCD recess. Comes with USB and cable and 3 ND filtersЗапросить расценки
- BeamGage StandardBeamGage Standard is our full-function software with an extensive set of ISO quantitative measurement, our patented UtraCal™ algorithm for the highest accuracy measurements in the industry.
- BeamGage ProfessionalBeamGage Professional has all of the functionality that BeamGage Standard includes. BeamGage Professional supports all of our beam profiling cameras, includes window partitioning to allow analysis of multiple beams on a single camera, and includes an automation interface written in .NET to push data to your custom applications.
What is the distance from the front of the camera to the sensor?
What is the saturation level of the SP928 camera?
What is the framerate of the SP928?
The effective frame rates listed in BeamGage specification sheets are the maximum rates typically achievable in actual use. Frame buffering, image processing techniques, graphical displays, and mathematical computation all add degrees of overhead to achieving higher frame rates. This can be further limited by the available PC hardware. BeamGage features two modes, Frame Priority and Results Priority, which change how the system balances the work. Results Priority acquires a frame, performs any enabled image processing, performs all calculations and updates the graphical displays before accepting another frame from the camera. This mode is most useful when a temporal sequence of frames is not necessary and should always be enabled when logging. Frame Priority mode will allow the calculations and graphical display updates to be interrupted if another frame is ready from the camera before those operations are complete. This can be useful when collecting all frames at the maximum camera frame rate is necessary.Close
What beam sizes can I measure with the SP920G?
44μm - 5.3mm The accurate beam size minimum is derived by the pixel size of the camera. In order to get an accurate measurement, there must be enough coverage of pixels to ensure that illuminating another pixel will not over exaggerate the beam size.
Follow this link to find out more.
VideosVideo Series: BeamGage Tutorials Video Series: BeamGage Tutorials
Is your laser's beam profile shaped correctly for your application?
This video teaches the fundamentals of laser beam profiles and discusses the benefits of profiling your laser beam.
Several case studies are presented showing before and after laser beam profiles.
This step-by-step tutorial will show you how to set up a camera-based beam profiling system on an industrial single-pulse laser welding system.
It will also demonstrate for you how to simultaneously analyze the laser's focused spot, measure the laser's energy per pulse, and measure its temporal pulse shape.
BeamMaker helps engineers, technicians, and researchers understand a beam's modal content by subtracting theoretically generated modes from real beam measurement data. Derive a perfect beam profile by specifying the mode, size, width, height, intensity, angle, and noise content - then comparing it to theoretically derived measurements. The end result is knowledge about how much the real beam varies from the desired beam.
Watch the BeamGage Tutuorials, including tips on handling your CCD camera, software install, introduction to the BeamGage user interface, the context-sensitive help system and user manual, customizing your reporting environment, and configuring BeamGage to display specific laser measurements.
Apples to Apples: Which Camera Technologies Work Best for Beam Profiling Applications, Part 1
Ensuring Quality Welds in Medical Devices
Why is Test Equipment Always Suspect?
The Optical Camera Trigger is an optical sensor that detects pulsed light sources and generates outputs to trigger a camera. The front aperture of the Optical Trigger must be directed at a light source that provides the necessary properties for trigger activation. (e.g. a laser flash lamp, a pick-off source from the main laser beam, or similar).