There is no time limit with the StarLab logging time – you can set ‘Unlimited’ logging in the log settings. (Note that, although StarLab does not impose any limits, there may be PC issues such as available disk space that could impose some upper limit.)
Power Meters FAQ's
Unfortunately the interface described above is not sufficient to establish the correct communication protocol for the computer USB to interface to the Nova II or Vega meter. The original USB cable (one is provided with each Nova II and Vega meter) is required for the StarLab software to communicate OK. The USB cable is P/N 7E01205 and is available to order.
Reinstall StarLab and it will fix this problem.
• RS232 Programming Commands The Ophir power meters that are supported through RS232 can be controlled remotely by a variety of programming languages. The programming and command information is available in the StarCom User Manual Appendix A, Guide to Programmers.
StarLab 3.50 does include new features which may require a firmware upgrade of the meter or PC interface, I.E. Juno, in order to operate with it. The required firmware is included with StarLab 3.50, but you do need to click on the More… link in the Select Device(s) menu in order to launch the Diagnostics menu and then proceed with the Upgrade firmware procedure. After performing the firmware upgrade, the meter or PC interface will connect with StarLab and operate normally. Note; Upgrading the firmware will not affect calibration.
The big difference between StarLab 3.40 and 3.31 is the enhanced support for our new Centauri dual channel meter. (Go download it free now and you'll see what I mean.) There are two other major improvements that I should mention:
- Added Low freq Power (Aka VCSEL) support for: Centauri, StarBright, Juno, Juno + and EA-1
- Added Power from Pulse support for Centauri, StarBright, Juno and Juno +
Find out more (and download StarLab) on the product page.