M², or Beam Propagation Ratio, is a value that indicates how close a laser is to being a single mode TEM00 beam, which in turn determines how small a beam waist can be focused. For the perfect Gaussian TEM00 condition the M² equals 1. M² cannot be determined from a single beam profile measurement. The ISO/DIS 11146 requires that M² be calculated from a series of measurements. M² is measured on real beams by focusing the beam with a fixed position lens of known focal length, and then measuring the characteristics of the artificially created beam waist and divergence. We have a number of solutions for the measurement of M² ranging from simple manual processes to fully automated dedicated instruments, depending on the frequency of the need to measure M² of lasers and laser systems.
This scanning slit M² measurement system accurately analyzes lasers with wavelengths from UV to Far Infrared with its silicon, germanium, or pyroelectric head. It features a compact portable design, immediate results, ISO compliant measurements, and operates in CW or kHz Pulsed modes which makes it ideal for comprehensive analysis of lasers of most wavelengths. Scanhead Travel: 500mm USB 2.0Interface M² Analysis Software includedShow Product >