The CO2 Laser Profiler Kits are designed to be conveniently placed in a horizontal beam, or under a down directed beam, to measure raw beam characteristics & stability. They consist of an A/R coated reflecting wedge, CaF2 filters, beam telescope to reduce larger beams, Gigabit Ethernet Pyrocam camera and BeamGage standard software.
- Beam sizes from 1mm-16mm in diameter
- 1 mm - 16 mm (depending on kit)
- Pyroelectric Array
- GigE Ethernet
- 1W - 500W
The LPK-CO2 Laser Profiler Kits are available with the following configurations.
Please note: Maximum beam size assumes zero diffraction from the wings of the beam. Beams of up to 1.5X the maximum size can be applied with minimal diffraction. Beams of up to 2X the above size can be applied, but noticeable diffraction will occur.
PY-III-HR-C-A BeamGage Standard software, Pyrocam IIIHR camera, LBS-100-IR-0.5 beam splitter/attenuator, base plate, miscellaneous hardware. Suitable for beams 1.0 to 6.4mm.$0.00Out of stock
PY-III-HR-C-A BeamGage Standard software, Pyrocam IIIHR camera, LBS-100-IR-5.0 beam splitter/attenuator, base plate, miscellaneous hardware. Suitable for beams 1.0 to 6.4mm.$0.00Out of stock
PY-III-HR-C-A BeamGage Standard software, Pyrocam IIIHR camera, LBS-100-IR-0.5 beam splitter/attenuator, 3X telescope, base plate, miscellaneous hardware. Suitable for beams 3.0 to 16mm.$0.00Out of stock
PY-III-HR-C-A BeamGage Standard software, Pyrocam IIIHR camera, LBS-100-IR-5.0 beam splitter/attenuator, 3X telescope, base plate, miscellaneous hardware. Suitable for beams 3.0 to 16mm.$0.00Out of stock
- BeamGage StandardBeamGage Standard is our full-function software with an extensive set of ISO quantitative measurement, our patented UtraCal™ algorithm for the highest accuracy measurements in the industry.
- BeamGage ProfessionalBeamGage Professional has all of the functionality that BeamGage Standard includes. BeamGage Professional supports all of our beam profiling cameras, includes window partitioning to allow analysis of multiple beams on a single camera, and includes an automation interface written in .NET to push data to your custom applications.
What is the dynamic range with the Pyrocam III, CCD's, or various other cameras?
- CCD camera manufacturers typically quote a signal-to-noise ratio of 50 to 60db. This refers to the peak signal before saturation divided by the RMS noise. This is a 20 log function, so 60db would be equivalent to a dynamic range of 1000, whereas 50db would be a dynamic range of about 300. However, it should be noted that this is comparing peak signal to RMS noise. The peak-to-peak noise is about 6 times the RMS noise. Therefore, the dynamic range in terms of peak signal to peak-to-peak noise ranges from about 50 to 180. With 8-bit digitizers, i.e., 256 counts, this means that the bottom 2-5 counts in the digitizer are noisy.
- The Pyrocam III pyroelectric camera also has a 60db signal-to-noise ratio. However, this is a worst case pessimistic specification. Typically the dynamic range is about 70db, which means the dynamic range is about 500 relative to peak-to-peak noise. This makes the camera very useful with 10 and 12-bit digitizers
How do I get the Pyrocam III to be recognized with BeamGage software?
Note; this answer applies to new Pyrocam III's that are shipped with BeamGage software or have been upgraded to work with BeamGage software. If a Pyrocam III is not licensed to work with BeamGage software, then the license needs to purchased and the Pyrocam III will need to be upgraded first.
The Pyrocam III uses the 1394a 400MBS protocol, but will operate as a 1394b 800MBS interface card, however it will revert still to the 1394a 400MBS protocol.
The Pyrocam III with BeamGage also uses a separately installed device driver which may not install and operate correctly at first.
Note for Windows 7. Windows 7 contains a new set of 1394 OHCI Compliant Host Controller drivers that will not communicate with the Pyrocam III. When the Pyrocam III is plugged into a Windows 7 computer nothing happens. No indication can be found that the Pyrocam III is plugged into the computer. The problem lies with the new OHCI Compliant Host Controller driver.
The solution is to replace the default "Texas Instruments 1394 OHCI Compliant Host Controller" with the "1394 OHCI Compliant Host Controller (Legacy)".
To replace the default Texas Instruments 1394 OHCI Compliant Host Controller:
- Open up Control Panel. Switch the View by in the upper right corner to Large Icons. Click on Device Manager.
- Click the arrow next to "IEEE 1394 Bus host controllers".
- Right-click on the 1394 port you wish to "fix", select Update Driver…
- The Hardware Update Wizard is displayed. Select "Browse my computer for driver software" .
- Select "Let me pick from a list of device drivers on my computer".
- Select "1394 OHCI Compliant Host Controller (Legacy)". Click the Next> button.
- When the Microsoft driver is installed, click the Close button.
The Pyrocam III Camera Driver Installation version 1.0 is available from our web address;http://www.ophiropt.com/laser-measurement-instruments/beam-profilers/services/software-download
We recommend that you right mouse click on the Setup.exe and "Run as Administrator".
If the Pyrocam III Camera Driver Installation version 1.0 doesn't work to install the Pyrocam3Wdf.sys driver properly, you may have to do the following;
Go to the Device Manager and with the Pyrocam III connected you should see it (probably) listed under Imaging Devices with a yellow exclamation error mark. Right click on it and select Update Driver Software…
Do not install automatically. Do install from a list. Do not search, but do install from a specific location.
Select Browse my computer for driver software and then select Browse to C:\Program Files\Spiricon\PyrocamKMDF\ and it will install the correct driver from there.
When or is the Pyrocam III going to get higher resolution?
The Pyrocam™ IIIHR & Pyrocam IV series with higher resolution are now available. For more information please access the webpage at the link below;
Illegal System DLL Relocation error with Windows Update and Pyrocam III
There is a Microsoft Windows XP update (KB925902) that causes an Illegal System DLL Relocation error with the Pyrocam III Control Console. Microsoft has released a hot fix for the problem with the RealTek Audio, and the fix also works for The Pyrocam III Control Console.
and scroll down to "Method 3: Install update 935448 from the Microsoft Download Center". Click the "Download the 935448 package now" link and follow the instructions.
VideosGetting Started with BeamGage Getting Started with BeamGage
In this video, we demonstrate how to configure BeamGage to display power and energy measurements from an attached Ophir sensor.If you can't see the video please click here
This step-by-step tutorial will show you how to set up a camera-based beam profiling system on an industrial single-pulse laser welding system.
It will also demonstrate for you how to simultaneously analyze the laser's focused spot, measure the laser's energy per pulse, and measure its temporal pulse shape.
BeamGage Professional and BeamGage Enterprise allow programmers to access all the functionality of the graphical user interface through LabView, Visual Basic, C++ and C#. This video is a short introduction to automation with a LabView demonstration.If you can't see the video please click here
Watch the BeamGage Tutuorial, including tips on handling your CCD camera, software install, introduction to the BeamGage user interface, the context-sensitive help system and user manual, customizing your reporting environment, and configuring BeamGage to display specific laser measurements.
Attenuation-Induced Error Due to Thermal Lensing in Beam Measurement
Stacking ND Filters: The Right Way and the Wrong Way
LBS-100 filter setSP90141
Replacement filter set$0.00Out of stock
LBS-100 -YAG filter setSP90142
Replacement filter set$0.00Out of stock
LBS-100 to L11058/L11059 adaptorSP90196
Mount L11058/L11059 camera to LBS-100 attenuator$0.00Out of stock
LBS-100 to 4X beam reducer adapterSPZ17029
This adapter enables mounting of the LBS-100 beam splitter/attenuator assembly in front of the 4X beam reducer. The combined assembly can image large high power beams in one unit.$0.00Out of stock